|No.333 Hezuo Road, Hi-Tech Zone, Chengdu 611731 CNemail@example.com|
The analyzer requires AC 220V，50Hz.
What if I need after-sales service?
If you meet difficulty when operating the analyzer, or have any questions about the SD1 analyzer, please feel free to contact the after-sales service via E-mail firstname.lastname@example.org , or telephone +86-028 69760308.
What is the requirements of the working environment of the SD1 analyzer?
ambient temperature: 10 ° C ~ 30 ° C.
relative humidity: 30% ~ 70%. atmospheric pressure: 860 hPa ~ 1060 hPa.
Please place the analyzer on a clean, level surface when operating.
What should be noticed before installing the analyzer?
when you receive the analyzer, you should check wheather the sealing of the packaing box is intact or not before unpacking. And check whether the analyzer is in good condition.
How to install the SD1 analyzer?
Take the instrument out from shipping carton and place it on a stable worktable;
Make sure appearance is intact and start to install
Connect the line Pin with the power supply
Press the power switch at the back of the instrument, the prompt light will on in the bottom of screen. The system starts and enters the interface to be tested.
How to install external printer?
The instrument is compatible with the printer which is HP PCL3 GUI printer language, like HP Deskjet 1010.
What should I do when the SD1 analyzer can't be turned on normally?
Please check 3 things:
check if the analyzer is powered
check if the power plug is loose
check the access voltage
What if the external printer can't work?
check whether system setting is under external printer
check whether printer USB line is connected well
check whether the printer model meets the machine requirements
What if the built-in thermal printer can’t work?
check whether system setting is under built-in printer
check whether printing paper is installed
What should I do when the results are abnormal?
check whether disc is expired, never use expired or damaged disc
check test assay setting.
When cannot enter the test interface？
Try to restart the analyzer.
what should I do when the touch panel can`t work?
you can connect mouse to use temporarily, and contact the local distributors or manufacturers later.
Contact Person: Mr. Bob Zhu